<!--
 * @Author: cgl
 * @Date: 2023-04-18 15:12:26
 * @LastEditors: cgl
 * @LastEditTime: 2023-09-04 15:23:28
 * @FilePath: \public-health-emergencies-fed:\新放射\新radiation-hygiene-fe\src\views\person\itemControl\item12.vue
 * @Description:
 *
 * Copyright (c) 2023 by cgl/微盛信息产业有限公司, All Rights Reserved.
-->
<template>
	<div>
		<table border="1" cellspacing="0" cellpadding="0" align="center" style="width: 100%;">
      <thead>
        <tr class="title">
          <td rowspan="2">序<br/>号</td>
          <td rowspan="2" colspan="2">检测项目</td>
          <td rowspan="2">检测方法<br/>与评价要<br/>求的依据<br/>条款</td>
          <td rowspan="2">技术要求</td>
          <td colspan="2">验收检测</td>
          <td rowspan="2">状态<br/>检测</td>
          <td colspan="3">稳定性检测</td>
        </tr>
        <tr class="title">
          <td>应<br/>检</td>
          <td>推<br/>荐</td>
          <td>应<br/>检</td>
          <td>推<br/>荐</td>
          <td>周期</td>
        </tr>
      </thead>
      <div class="tableBody">
        <tbody>
          <tr>
            <td>1</td>
            <td colspan="2">剂量偏差</td>
            <td>6.1.1</td>
            <td>≤3%</td>
            <td>√</td>
            <td>—</td>
            <td>√</td>
            <td>√</td>
            <td>—</td>
            <td>一周</td>
          </tr>
          <tr>
            <td>2</td>
            <td colspan="2">重复性（剂量）</td>
            <td>6.1.2</td>
            <td>≤0.5%</td>
            <td>√</td>
            <td>—</td>
            <td>√</td>
            <td>√</td>
            <td>—</td>
            <td>六个月</td>
          </tr>
          <tr>
            <td rowspan="2">3</td>
            <td rowspan="2">线性</td>
            <td>剂量</td>
            <td>6.1.3.2</td>
            <td>≤2%</td>
            <td>√</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
          </tr>
          <tr>
            <td>剂量率</td>
            <td>6.1.3.3</td>
            <td>≤2%</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
          </tr>
          <tr>
            <td>4</td>
            <td colspan="2">随设备角度位置的变化（剂量）</td>
            <td>6.1.4 </td>
            <td>≤3%</td>
            <td>—</td>
            <td>—</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
          </tr>
          <tr>
            <td rowspan="2">5</td>
            <td rowspan="2">随机架旋 <br/> 转的变化<br/>（剂量）</td>
            <td>X</td>
            <td>6.1.5</td>
            <td>≤3% </td>
            <td>—</td>
            <td>—</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
          </tr>
          <tr>
            <td>电子</td>
            <td>6.1.5</td>
            <td>≤2%</td>
            <td>—</td>
            <td>—</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
          </tr>
          <tr>
            <td>6</td>
            <td colspan="2">日稳定性（剂量）</td>
            <td>6.1.6 </td>
            <td>≤2%</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>六个月</td>
          </tr>
          <tr>
            <td>7</td>
            <td colspan="2">X射线深度吸收剂量特性</td>
            <td>6.2.1 </td>
            <td>≤3%或≤3 mm</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>六个月</td>
          </tr>
          <tr>
            <td>8</td>
            <td colspan="2">电子线深度吸收剂量特性</td>
            <td>6.2.2 </td>
            <td>≤3%或≤2 mm</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>六个月</td>
          </tr>
          <tr>
            <td rowspan="2">9</td>
            <td rowspan="2">X射线方<br/>形照射野<br/>的均整度</td>
            <td>5 cm×5 cm<br/>~30 cm×30 cm </td>
            <td>6.3.1 </td>
            <td>≤106%</td>
            <td>√</td>
            <td>—</td>
            <td>√</td>
            <td>√</td>
            <td>—</td>
            <td>三个月</td>
          </tr>
          <tr>
            <td>大于30 cm×30 cm </td>
            <td>6.3.1 </td>
            <td>≤110% </td>
            <td>√</td>
            <td>—</td>
            <td>√</td>
            <td>√</td>
            <td>—</td>
            <td>三个月</td>
          </tr>
          <tr>
            <td>10</td>
            <td colspan="2">X射线方形照射野的对称性 </td>
            <td>6.3.1 </td>
            <td>≤103%</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>六个月</td>
          </tr>
          <tr>
            <td rowspan="3">11</td>
            <td rowspan="3">电子线照<br/>射野的均<br/>整度</td>
            <td >沿两主轴方向上<br/>80%等剂量线</td>
            <td>6.3.2</td>
            <td>≤15 mm</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>三个月</td>
          </tr>
          <tr>
            <td >沿两主轴方向上<br/>90%等剂量线</td>
            <td>6.3.2</td>
            <td>≤10 mm</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>三个月</td>
          </tr>
          <tr>
            <td >两对角线上<br/>90%等剂量线</td>
            <td>6.3.2</td>
            <td>≤20 mm</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>三个月</td>
          </tr>
          <tr>
            <td>12</td>
            <td colspan="2">电子线照射野的对称性</td>
            <td>6.3.2</td>
            <td>≤105%</td>
            <td>√</td>
            <td>—</td>
            <td>√</td>
            <td>√</td>
            <td>—</td>
            <td>三个月</td>
          </tr>
          <tr>
            <td>13</td>
            <td colspan="2">照射野的半影</td>
            <td>6.3.3</td>
            <td>应符合厂家给出值</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
          </tr>
          <tr>
            <td rowspan="2">14</td>
            <td rowspan="2">照射野的数字<br/>指示（单元限<br/>束）</td>
            <td>5 cm×5 cm<br/>~20 cm×20 cm</td>
            <td>6.4.1</td>
            <td>≤3 mm或≤<br/>1.5%</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td>大于20 cm×20 cm</td>
            <td>6.4.1</td>
            <td>≤5 mm或≤ <br/>1.5% </td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td rowspan="2">15</td>
            <td rowspan="2">照射野的数字<br/>指示（多元限<br/>束）</td>
            <td>10 cm×10 cm</td>
            <td>6.4.1</td>
            <td>≤3 mm </td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td>最大照射野</td>
            <td>6.4.1</td>
            <td>≤5 mm或≤<br/>1.5% </td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td >16</td>
            <td colspan="2">辐射束轴在患者入射表面上的位置<br/>指示 </td>
            <td>6.4.2</td>
            <td>≤2 mm</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td >17</td>
            <td colspan="2">辐射束轴相对于等中心点的偏移</td>
            <td>6.5.1</td>
            <td>≤2 mm</td>
            <td>√</td>
            <td>—</td>
            <td>√</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td >18</td>
            <td colspan="2">等中心的指示（激光灯）</td>
            <td>6.5.2</td>
            <td>≤2 mm</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一天</td>
          </tr>
          <tr>
            <td rowspan="4">19</td>
            <td rowspan="4">旋转运动标尺的<br/>零刻度位置</td>
            <td >机架旋转轴</td>
            <td>6.6.1</td>
            <td>≤0.5°</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td >限束系统旋转<br/>轴</td>
            <td>6.6.2</td>
            <td>≤0.5°</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td >治疗床面纵向<br/>转动轴</td>
            <td>6.6.3</td>
            <td>≤0.5°</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td >治疗床面横向<br/>转动轴 </td>
            <td>6.6.3</td>
            <td>≤0.5°</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
          <tr>
            <td rowspan="3">20</td>
            <td rowspan="3">治疗床的运动精<br/>度</td>
            <td >垂直</td>
            <td>6.7.1</td>
            <td>≤2 mm </td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>六个月</td>
          </tr>
          <tr>
            <td >横向</td>
            <td>6.7.2</td>
            <td>≤2 mm </td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>六个月</td>
          </tr>
          <tr>
            <td >前后</td>
            <td>6.7.3</td>
            <td>≤2 mm </td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>六个月</td>
          </tr>
          <tr>
            <td rowspan="3">21</td>
            <td rowspan="3">治疗床的刚度</td>
            <td >纵向（高度的<br/>变化）</td>
            <td>6.8.1</td>
            <td>≤5 mm</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一年</td>
          </tr>
          <tr>
            <td >横向（侧向倾<br/>斜角度）</td>
            <td>6.8.2</td>
            <td>≤0.5°</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一年</td>
          </tr>
          <tr>
            <td >横向（高度的<br/>变化）</td>
            <td>6.8.2</td>
            <td>≤5 mm </td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一年</td>
          </tr>
          <tr>
            <td >22</td>
            <td colspan="2">治疗床的等中心旋转</td>
            <td>6.9</td>
            <td>≤2 mm</td>
            <td>√</td>
            <td>—</td>
            <td>—</td>
            <td>√</td>
            <td>—</td>
            <td>一个月</td>
          </tr>
        </tbody>
      </div>

      <tr>
        <td colspan="11" style="text-align: left;">注：“√”表示应进行对应项目的检测，“—”表示不进行对应项目的检测。</td>
      </tr>
		</table>
	</div>
</template>

<script>
export default {
  name: "item50",
  data() {
    return {
    };
  },
  created() {
  },
  methods: {
  }
};
</script>

<style lang="scss" scoped>
  @import './content.scss';
.tableBody {
  display:block;
  height:50vh;
  overflow-y:scroll;
  scrollbar-width: none;
}

tbody,table thead {
  display:table;
  width:100%;
  table-layout:fixed;
}
tbody, td{
  border-collapse: collapse;
  border: 1px solid #606266;
}
</style>
